Station Programmer Installation Software:
Intech Micro Remote Station Programmer software.
This software supports the programming of stations via COM/USB ports or Ethernet network and the reading of SD cards on the 2400-A16-SD.
- The Station Programmer software requires Microsoft .NET Framework 2.0 to be installed first
(Microsoft .NET 3.5 SP1, which includes .NET 2.0, can be downloaded free).
- Windows users:
- Installation requires the user to be an administrator.
- Desktop shortcut icon will only be installed to the currently logged on user.
- Extract all files to a folder.
- Right click on setup.exe and click 'run as administrator'.
- If using via the XU-USB programming key (2400-XX stations, USB programming port):
- Ensure that the Station Programmer software is installed before plugging in the XU-USB as this ensures that the required USB driver is installed.
- When upgrading from a previous version - please disconnect the XU-USB before installing.
- Plug in the XU-USB to the computer (Standard size USB port), then the 2400-XX station. Then power up (ensure that the voltage level matches the 2400-XX stations' label), the 2400-XX station.
- Note: 2400-XX stations can also be programmed via one of their Comms ports.
- If using via Network (Ethernet TCP/IP) -NET stations:
- You will need to allow the Station Programmer software access through the Windows Firewall in order to find the Ethernet converter (XPort®) and set the IP address etc. When the Windows Security Alert window appears, click 'Allow access'.
- Factory set IP address = 192.168.1.100
- Factory set Port = 10001
- This software is to be installed and/or setup by the Installer with Network IT support staff. Please note that Intech will not be able to help with the setup and operation of an Ethernet TCP/IP network and assigning of IP addresses. As part of commissioning IT support staff should assign a static address to each 2400-A16-NET, 2100-A16-NET or 2100-NET. The 2400-A16-NET, 2100-A16-NET or 2100-NET should be listed as part of the Ethernet system for support by the IT support staff. It is considered good practice to write the assigned IP address/port numbers onto a label and attach to the unit for easy future reference.
- If programming the 2100-A16 or 2100-A16-NET for Modbus applications:
- Leave the 2100-A16 or 2100-A16-NET in MicroScan ASCII mode (factory default). Once all channels etc are programmed, follow the steps to change over to Modbus mode.
- Note: The 2400-A16, 2400-A16-NET and 2400-M-R will auto-detect Modbus operation.
- If using SD memory data logging with the 2400-A16-SD:
- Configure the 2400-A16-SD Input Ranges etc as usual.
- Configure the SD recording parameters.
- Configure the Pages & Lines for all Inputs being used.
- Initialise SD card if more than one computer is used for download.
- Note: Using serial comms to manually download data is very slow, and is not recommended.
WARNING: Ensure that all programming is carried out BEFORE any wiring takes place!
Download Now - 7.6MB - April 2, 2015 - (Ver 5.2.3613 - reports as 5.1.2 under Windows Installed Programs)
XU-USB programming key (2400-XX stations, USB programming port)
Note: 2400-XX stations can also be programmed via one of their Comms ports.